When the regeneration is started, a pop-up window appears with the question "Are you sure you want to regenerate? The regeneration can be aborted with "NO". If you have selected "YES", you will subsequently be prompted to check the regenerating gas flow.
From this point on, you can no longer completely interrupt the regeneration and circulate it again via the Reactor Copper Molecularsieve (RKM). Depending on the software version, the complete regeneration runs through or the next step to abort the regeneration is started. You can read the remaining regeneration time on the "Status" page in the regeneration gas menu.
Even if the replenishment gas cylinder does not contain the required quantity of replenishment gas or is empty, you must allow the replenishment process to run through.
The system will not be damaged if a regeneration without gas has been started accidentally. The regeneration cycle must be completed, otherwise the corresponding Reactor Copper Molecularsieve (RKM) will be locked.
If possible, the glovebox should be left powered on and circulating (please see our previous post). However, if you are required to power down the system do as follow:
If your Oxygen Analyzer does show too high ppm values or does not change the displayed value at all. What might be wrong?
Please check if you do have higher concentrations of volatile, Oxygen-containing, or corrosive substances in the glovebox?
Please make sure you do not have any leakage in the box. Did you check your gloves for holes?
If your Moisture Analyzer does not change the displayed value at all � frozen value. What might be wrong?
Please check if your analyser head is dirty or completely dry (no phosphoric acid left).
You do not know how? Please check out our How-To video to clean and get phosphoric acid back on the sensor head.
If your Solvent Analyzer does not change the displayed value at all � shows permanent 0 ppm. What might be wrong?
Please check if your analyser is still full functional and place an alcohol pad in front of the analyser head. An immediate change of the ppm value should be visible.
You do not have a pad please ask at firstname.lastname@example.org for a free sample or order (item-no. 2612298).